Simple Usage Information
Press the 'Power' button to switch the device on and off.
The switch on the side of the device is for selecting between menus.
The '' button is for changing options on the screen.
The 'Back' button is used to display the previous displayed value.
How to use.?
1: Use the POWER switch to turn the device on and off.
2: The device is several cm from the plant root. position. Too much immersion in the bottom of the device or too much immersion affects the measurement. In order to perform the correct measurement, the instrument probe should be inserted into the ground between 1/2 and 3/4.
3: When dipping the probe into the ground, you will see the values on the display. The reason is that the soil moisture is not homogeneous. Humidity may vary depending on the conditions. Certain types of soils tend to form moisture communities. It is recommended to take at least 2 different regions to make sure that the measurement accuracy is correct.
4: It takes 4-6 seconds for the value on the LCD display to settle.
5: Remove the device from the ground after measurement.
6: Clean the instrument probe with a clean, soft cloth and prepare for the next measurement.
* The device is developed and manufactured for instant reading.
* Do not leave it for a long time to remain damp.
* Do not leave in the soil with the handle.
* Automatically turns off after 2 minutes when the device is not active. (Press the POWER button to switch it back on again.)
There are moisture values of many popular plants in the memory of the device.
You can compare the difference between the measurements during the measurement by selecting the plant you grow from the screen.
SAMPLE: We're measuring for beans. Our device has measured as 4, but if the value to be on the screen appears to be 2-3, then we understand that we should not water.
Note: Plant options on the device screen are in English. A table showing the Turkish equivalents of these options will be sent with the device ...